3 liter dental autoclave
Power: 110-240v, 500w (with over heat protection, safety valve)
Sterilization temperature: 135ºC, 0.22Mpa
Sterilization pressure: 0.22Mp
quick sterilizing within 15 minutes
Sterilizar volume: 3L
chamber size: ø125*240mm
Required water: distilled water or purified water
carton size: 370*260*430
1. Model: SUN3-I
2. Inner chamber of the quick steam sterilizer (Diameter X Depth) 125mm X 195mm
3. Rated Voltage: 220V±22V,50Hz±1Hz ; 110V±11V,60Hz±1Hz
4. Rated Power: 500VA
5. Sterilizing:(Pressure/Temp.): 200~230Kpa/134ºC
(Both of above are Under Standard Atmospheric pressure)
6. Fuse-Melting pipe: T15A;T30A
7. Water Consumption volume per recycling round: 0.15L
8. Operating Temp: 0-40oC
9. Net weight:7.5kg
10. Gross Weight:7.75kg
11. Appearance size:30*19*35cm
12. Package Size:38*29*45cm
14. Sterilization time: 11min
15. Sterilizar Volume: 3 Litres
16. Required Water: Distilled water or purified water
17. SKU: MT-D-SUN3(I)-220
Maintenance and Service
1. Clean the distilled water tank by medical once a month.
2. Clean the interior surface of the chamber by alcohol once a month.
3. Replace the fuse.
(1) Shut off the power
(2) Revolve the fuse base by counter-clockwise by screwdriver to bring out the melting fuse to be replaced;
(3) Replace the new fuse and reset the fuse base on the original position, then use the screwdriver to fasten the base by clockwise.
(4)Check the correctness of the parameter of the new fuse replaced.
4. Clean the sealing ring periodically
The user shall clean the sealing ring periodically for avoiding the influence to the seal caused by the dust/dirt left due to the long period use. Use the smooth cloth with distilled water to wipe the surface of the sealing ring or sealing cap gently. The user shall discharge the sealing ring for further cleaning or replacement if the air-leaking problem can not be resolved after the above process (The user shall discharge the sealing ring periodically for cleaning and inspection as recommendation by the supplier).